# 표준전자식 혈압계와 수은혈압계, 전자혈압계의 비교 선행연구

• Yahng, J.S. (Center for medical measurements, Korea Research Institute of Standards and Science) ;
• Lim, H.K. (Center for medical measurements, Korea Research Institute of Standards and Science) ;
• Cho, D.H. (Dept. of admissions officer, Dankook University)
• 양지상 (한국표준과학연구원 의료융합측정연구단) ;
• 임현균 (한국표준과학연구원 의료융합측정연구단) ;
• 조동헌 (단국대학교 입학사정관팀)
• Accepted : 2012.09.20
• Published : 2012.09.30

#### Abstract

Devices to measure the blood pressure of patients are being used without any calibration in a hospital. It is an important to show consistent values when any medical devices measure the same patients regardless they are sphygmomanometer or fully automatic electronic blood pressure meter. We compared sphygmomanometer and fully automatic electronic blood pressure meters with standard digital blood pressure monitor (SDBPM) to evaluate the consistency of the small healthy subjects. We measured the blood pressure from six healthy subjects (three of 20~40 years and three of 40~60 years old). Two sphygmomanometer and two fully automatic electronic blood pressure meters were used and compared with the SDBPM. Blood pressures measured from right and left arms each and were compared. All six healthy subjects showed normal blood pressure values. In general, left blood pressure values showed higher values than right side. Comparing SDBPM, with the other monitors, the systolic pressure showed ${\pm}$ 34.8% difference and ${\pm}$ 33.3% for the diastolic pressure. Correlation between SDBPM and Sphygmomanometer was 0.59~0.71, and 0.50~0.70 for fully automated digital BP monitors. It fell in grade-D when we apply the BHS(British hypertension society). AAMI(American association for the advancement of medical instrumentation) also showed unsatisfactory results for the mean value (${\leq}$ 5 mmHg) and standard deviation (${\leq}$ 8 mmHg). We tested sphygmomanometer and fully automatic electronic blood pressure meters and compared with a standard digital blood pressure monitor. All devices showed inconsistent blood pressures. A reliable calibration system is highly needed for all devices in all hospitals.

#### References

1. C. G. Son, 2009 patient survey In-depth Analysis, Ministry of Health and Welfare, 2011.
2. K. Kang, D. W. Kim, D. W. Kim, H. C. Kim, K. S. Park, Medical Instrumentation Engineering, Yeomungak, Seoul Korea, pp. 395-449, 2012.
3. L. A. Taylor, R. G. Bennett and T. J. Dorsett, "The application of signal averaging to the NIBP oscillometricwaveform", Proceedings of Annual International Conference of the IEEE EMBS, vol. 1, pp. 691-2, 1995
4. C. T. Lin, S. H. Liu, J. J. Wang, and Z. C. Wen, "Reduction of interference in oscillometric arteiral blood pressure measurement using fuzzy logic", IEEE transactions on Biomedical Engineering, vol. 50, no. 4, pp. 432-41, 2003. https://doi.org/10.1109/TBME.2003.809502
5. C. Huo, J. Wang, J. Zhang, and S. Liu, "Blood pressure measurement based on wavelet analysis and fuzzy inference method", Proceedings of the 6th World Congress on Intelligent Control and Automation, vol. 1, pp. 4382-6, 2006
6. T. G. Pickering, "What will replace the mercury sphygmomanometer?" Blood Press Mount, vol. 8, pp. 23-5, 2003. https://doi.org/10.1097/00126097-200302000-00005
7. C. G. Lee, J. W. Park, J. S. Moon, N. S. Kim, and K. H. Kim, "Age effects on the differences between Omron HEM 907 blood pressure monitor and manual mercury sphygmanometer blood pressure readings", The Korean Society of Hypertension, vol. 17, no. 3, pp. 114-24, 2011. https://doi.org/10.5646/jksh.2011.17.3.114
8. J. N. Choo, "Clinical Evaluation of the Accuracy of Electronic Home Blood Pressure Measuring Devices", The Korean journal of fundamentals of nursing, vol. 9, no. 1, pp. 101-12, 2002.
9. E. O'Brien, "Replacing the mercury sphygmomanometer : Requres clinicians to demend better automated devices", Br Med. J. vol. 320, pp. 815-816, 2000. https://doi.org/10.1136/bmj.320.7238.815
10. American association for the advancement of medical instrumentation, American national standard for electronic or automated sphygmomanometers, Arlington Virginia, 1987.